K7L-AT50 Liquid Leakage Sensor Amplifier/Liquid Leakage Sensor Amplifier with Disconnection Detection Function
Rated power supply
voltage12 to 24 VDC (Allowable voltage fluctuation range: 10 to 30 VDC)Operate resistanceΩ to 50 MΩ, variable
Range 0: 0 to 250 kΩ. Range 1: 0 to 600 kΩ
Range 2: 0 to 5 kΩ. Range 3: 0 to 50 kΩ
The range is set using the DIP switch on the side of the Sensor Amplifier. (Refer to DIP Switch
Set the corresponding pin of the DIP switch in the up position. (For range 0, set all 3 pins in the
down position.) The adjuster (ADJUST) on the top of the Sensor Amplifier sets the resistance
value for detection within the set range. It is factory-set to the upper limit. (Normally, the K7L
can be used with the adjuster at this setting.) With any range, resistance values can be set from
*Detection signal: 10 VDC max., 200 ms
Detection time: 10 s max.
Recovery: Operation is recovered by resetting the power supply.Release resistance105% min. of operate resistanceOutput
configurationOpen-collector transistor output with 100 mA at 30 VDC max. for both liquid leakage detection
and disconnection detection.
Max. 30 VDC, 100mA
If the rightmost pin of the DIP switch on the side of the Sensor Amplifier is set to the down
position, the output turns ON when liquid is detected; if it is set to the up position, the output
turns OFF when liquid is detected.Wiring distanceConnecting cable: 50 m max.
Sensing Band length: 10 m max.
These values are possible on condition that a completely insulated 3-conductor VCT cable with a
thickness of 0.75 mm2 and a dielectric strength of 600 V is used together with a Liquid Sensing
Band specified by OMRON. (A 0.2-mm2 cable can also be used.)Accessories Terminal BlockScrewdriver for ADJUSTTerminatorK7L-AT50/AT50B11—K7L-AT50D/AT50DB/